I had some issues with system settings like delimiter, comma and dots for csv files. Message 6 of 9. The spectrum part is not very intelligent but its ok for now. Cannot open include file: Now, I want a computer interface to be able to operate it.
|Date Added:||1 December 2009|
|File Size:||13.3 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
You could have states such as: I am stepping the temperature and sweeping voltage. That’s just index array, which gives you the 1-d array that’s relevant. Message 8 of 9. I have two modes of experiment. However adjusting Laekshore and I values are bit time consuming. Message 6 of 7. I am not sure how the works too F Casian 45 7.
Lake Shore Cryotronics 330 Controller
On error stop the timer and take the necessary steps to correct, report to the user. Let me explain the experiment.
When you really only want to plot data of the current sweep you need to delete the data stored in the shift registers aka feedback nodes when you start a new sweep! Message 5 of 7.
I am using Win10 and a 64bit system. Does anyone have an Idea how to do that? Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type.
Hello James Thanks for the reply.
io – LabVIEW VISA Read doesn’t read data after system settings change – Stack Overflow
For learning about the basics of Lakesohre programming, here are some great resources for getting started. Each package of the data stream is as follows: I can find the instrument, but I cannot communicate with it because the pyvisa ‘write’ command gives an error when I try Derek Lim 93 Check if the timer set in the previous state has expired? Message 7 of 7. If so try to debug.
Also how do we know the best values of PID setting to run the experiment.?? In its manual it is specified something about the lakeshord terminal and I have some basic commands.
ResourceManager Traceback most recent call last: I am not sure if the spectrograph I have works continuously or not. I want to control a Keithley with an Ubuntu PC. The Keithley is connected to the computer via a a RSto Most Active Software Boards: Whenever I try to identify the device, it works just fine, both in reading and writing.
If you are trying to use the HIGH output for heater power to achieve fine control at low temperatures, you’re going to have a bad time. Likn Markn 23 1 6.
VISA will not deal with integers, as your example shows it will output a string. I am quite confused because integers have no decimals, commas or other symbols.